Home » MI 8000B Binary Voltage Divider
The MI 8000B Binary Voltage Divider is a highly versatile, accurate, self-balancing instrument that meets laboratory requirements for scaling between 10 Volt references or any voltage between 1 mV to 10 Volts. Automatic self-calibration ensures ratios to nine significant digits with linearity deviations of less than 0.02 ppm. The model 8000B has a 20-channel “built-in” scanner addressed individually via the Windows operating software for performing automatic measurements. Both hardware and software standard cell protection circuits are built-in.
In order to address the increasing demand for automated calibration of DCV ranges of most precise multi-function calibrators, as well as the linearity verification of the long scale DVM’s, the option of fully automated bipolar measurement has been developed. This allows users to do a measurement of ± 10 V range with 8000A and extend it to ± 1200 V when the 8001B extender is used.
The fully documented calibration of bipolar voltages at the output of the calibrator is traceable to the laboratory’s 10 V reference standard. Determining the 8000B correction factors and standardizing the source at both polarities gives additional confidence to calibration results.
The range of the MI 8000B Binary Voltage Divider can be extended to 1200 Volts with Measurements International’s precision divider extender (model 8001B). The latest development HW and SW features of 8000B allows fully automated bipolar measurements without manual intervention. This, in combination with model 8001B extender, brings full automation of DCV ranges calibration and linearity verification of multi-function calibrators and long scale DVM’s at a range up to ± 1200 V.
The principle of the 8000B Automatic Potentiometer is based on the Binary Voltage Divider (BVD). The BVD requires two voltage references and a DMM detector. The first reference or source is a low drift, stable, noise-free 10 Volt source which is connected to the rear on the 8000B source input. The most important thing about the source is its stability. The reference to the BVD is supplied from a calibrated stable voltage reference, MI model 1000A or Fluke model 732A or B. The source and 8000B are standardized against the calibrated reference for making absolute voltage measurements.
The DMM detector with an input impedance of 10 G or higher is then used to measure the difference between the output of the BVD and the voltage under test. An isolated guard circuit is provided to guard the BVD and the DMM detector when performing measurements. The guard voltage can also be used to drive the guards of the cell enclosures under test to reduce leakage problems.
Open the datasheet for detailed specifications of this product.
+31 252 - 621 080
"The ABB Power Grids Jumet organisation is an innovative development and production facility of ABB that manufactures active filters and battery energy storage systems to support a stable grid. To test these systems we have acquired two Cinergia GE/EL +120kVA eplus units capable of doing grid emulation (AC + DC), operating as an electronic load (AC +DC) and being a battery emulator. We have experienced very good support from TTMS in obtaining these two Cinergia units. This included organising an on-site demonstration and supplying loan systems prior to the delivery of our final systems. This way we could start testing as early as possible. TTMS also organised training for the operators from the original manufacturer. This support allowed us to introduce the first battery energy storage systems even prior to the full delivery of the Cinergia systems."
University of Luxembourg
“At the Energy and Environment department of the Faculty of Science, Technology and Medicine (FSTM) of the University of Luxembourg we do have different research projects related to green energy contributing to the development of eco-cities. As a long-standing partner of our research unit, TT&MS has proven that they are able to answer even the most complex questions in a professional manner. They have shown that they are always up to date with the latest technology and do not spare any effort to look at the application and even the situation on site. The exchange of ideas was always very pleasant and took place quickly. We look forward to further years of fruitful cooperation.”
Doelbewust Online Marketing